The dental unit is one of the necessary equipment in the dental clinic. Doctors may encounter some equipment problems when using the dental treatment machine. If these problems occur, many problems need to be considered.
For example, which dental unit manufacturer is better for maintenance? How long does it take for a home repair? How much will the service provider's on-site repair cost? Will it affect normal operating income for a long time? The outpatient team is too large, and the equipment is not managed and maintained?
Don't be afraid of these problems now, AJAX Medical Equipment Co., Ltd. will teach you the analysis and troubleshooting methods of the common problems of the dental unit, very fast repair and maintenance of the equipment, some small problems do not need to trouble the service provider at all, you can solve it by yourself!
1. The pipeline is connected incorrectly.
2. The pipeline is blocked by dirt.
3. The weak suction cup is blocked.
4. The weak suction main valve is blocked by dirt.
5. Poor water quality leads to blockage of pipes.
Remedy: replace the parts that cannot be repaired, check and repair the circuit of the dental unit; clean the weak suction main valve.
It is recommended that the user clean the dirt collection cup regularly (clean it once or twice a month), rinse it with a three-way gun, and put it back as it is, keeping the inside of the tube unobstructed.
1. The control wire and the motor wire are connected incorrectly.
2. The mask or controller is damaged.
3. The switch in the automatic direction is broken or the welding is wrong.
4. The foot switch is broken.
5. The program controller is out of power, check the fuse or power input
Exclusion method; check and connect the line of the dental unit, and change the mask.
1. The bulb is burnt out.
2. Poor contact of the lamp.
3. The secondary side of the power transformer is blown.
4. The wires inside the movable joint are twisted and broken.
5. The contact of the insertion is poor, and the connector of the dental unit is loose.
6. The public line does not penetrate into the controlled breaker of the dental unit.
7. Check whether the output of the transformer has voltage.
Remedy: change the bulb of the same model, turn off the power supply, check and connect the plug, replace the fuse of the same model, and replace the new line.
Dental units cannot be lacking in dental clinics. After all, a dental unit with equipment failure may affect the daily operation of the clinic.
AJAX Medical Equipment Co., Ltd. provides you with high-quality dental treatment machines and very high-quality after-sales service. We are very much looking forward to cooperating with you.