Designed with smart ergonomics, Ajax dental stool eases the pains caused by physical dental work by holding your posture and position in a health form. Taking different needs for the medical stool into account, Ajax provides three types of dental chairs for dentists consisting of the medical stool with wheels, the medical stool with back and the medical stool with handle (for example, D1 Doctor Stool With Wheels, N1 Nurse Stool and N4 Dental Assisting Chair). A wide range of products and a variety of color options, with ultimate comfortable upholstery, it can prolong your career life.
Your work can be demanding, we got your back!
It is recommended to clean and disinfect the waterlines of dental headpiece tubes, triplex syringe and scaler and mouthwash water tube used by the clinic with 50mg/L (ppm) slightly acidic hypochlorous acid disinfectant water.
1. Single bottle
Turn the water source changeover switch to the direction of tap water, take the water bottle and pour it into the prepared 1:3 micro-acid hypochlorous acid, and then turn the water source changeover switch to the direction of the water bottle. Pick up all the handpiece tubes, place triplex syringe over the cuspidor/sink, open the status valve, press the switch of foot pedal&control, rinse them continuously for 30 seconds, hang all the tubes back to the instrument tray and wait for 20 minutes; Switch the water source to the tap water, remove the water bottle, clean it, introduce distilled water and put it back in the dental chair, handpiece tubes, place triplex syringe over the cuspidor/sink, open the status valve, press the switch of foot pedal&control, rinse them continuously for 1 minute.
2. Double bottle (Please refer the video)
3. Automatic disinfection (An operation demonstration video is available)
It depends on different models, Each model of Ajax can record up to 3 positions. AJ25 can record up to 3 positions of 6 users.
It is forbidden to disinfect dental chairs by means of alcohol, fortified glutaraldehyde disinfectant, 84 disinfectant, benzalkonium bromide disinfectant and benzyl chloride containing ammonium propanol disinfectl chairus acid and alkaline disinfectants. The manufacturer shall assume no liability or responsibility for any damage of leather of dental chairs (such as color difference chap and aging) caused by it.
Ajax dental units are manufactured with very strict quality checks. The testing methods can effectively test the life, performance, and function of materials. Our dental units usually last from 15-20 years.
You can go to the "contact us" page, sent us the items you are interested along with the information required, we will reply within 24 hours. You can also reach to us on Facebook/Ajax Dental.